April 13, 2008 to April 18, 2008
DOI Bookmark: http://doi.ieeecomputersociety.org/10.1109/ICONS.2008.66
This paper presents design and assembly of mixed electronic circuitry for measured signal processing of the capacitive difference pressure sensor, as well as analysis of the obtained results. The smart pressure sensor provides values of measured pressure via 4 - 20 mA current loop output. The loop current is also used for sensor circuitry supplying. This means that current consumption of the whole sensor electronics should be less than 3.5 mA even in extended industrial temperature range from –40 to +125 ?C.
Steffan Pavel, Michal Pavlik, Radimir Vrba, "Smart Differential Pressure Sensor with Bluetooth Communication Interface", ICONS, 2008, 2008 3rd International Conference on Systems (ICONS '08), 2008 3rd International Conference on Systems (ICONS '08) 2008, pp. 363-367, doi:10.1109/ICONS.2008.66